Wash the 100 mL trays :
- For the µMB (micro Magnetic Beads) tray, if microbeads remain in the bottom, rince them with isopropanol 70% first.
- For all tray, wash at 55°C max
Autoclave all the following components:
I. The washing station and pipes
II. The bottle « liquid system » and the bottle « waste ».
Liquide system waste
The thermoshaker has a cooling liquid inside that have to be filled every month :
- unscrew the 2 screws
- Mont the needle with intermediate holder :
- Aspirate Inheco liquid from THERMOSHAKE COOLING LIQUID and fill it inside the Thermoshake until you can see it in the filling nozzle.